Džepne veličine, uz pouzdanost za laboratorije
Fresh capillary whole blood
Capillary: 6 uL
See test strip package insert for details
Reporting range: 0.8–8.0 INR. Results outside this range will be displayed as Out of Range.
At minimum, 640 patient results, 300 Liquid Quality Control results, 300 error messages
Xprecia™ Data Management Software (DMS)
Provides a simple, easy way to transfer data to and from the Xprecia Stride analyzer and a PC. The analyzer connects to a computer via USB to transmit:
Operator ID list; analyzer and operator settings
Analyzer software (firmware) upgrades
Patient and Liquid Quality Control results
Analyzer fault log
International Normalized Ratio (INR)
Electronic Quality Control (EQC)
Two quality-control checks verify the integrity of the test strips. The first control checks the presence of adequate sample and reagents on the test strip, and the second control detects test strip degradation due to exposure to environmental conditions. If either control fails, the system reports an error rather than an incorrect result.
Liquid Quality Control (LQC)
LQC tests help meet regulatory compliance requirements applicable to your facility. Use control solutions to perform quality-control checks on both the analyzer and test strips to ensure they are functioning correctly.